Portable Optical Power Meter H306
The portable optical power meter and red light source all-in-one machine is a new generation of measuring instruments developed through many years of design experience and can meet the current latest measuring requirements in the optical communication field. It can accurately measure the intensity of input optical signals on up to 6 strictly calibrated operating wavelengths. This makes this instrument become a strong tool on many application occasions such as the installation, operation and maintenance of optical fiber networks and the measurement experiment of factories and research units.
v High sensitivity and ultra-wide dynamic measuring range
The reference value can be set to perform REF relative measurement
v New function of user self-calibration
v Logarithm power (dBm) and linear power (xW) index measurement
v Automatic memorization of operating parameters set by the user
v Frequency identification function of input optical signals
v Automatic off function can be set
v Real-time display of battery level
v Backlight automatic off function is optional
v Universal interface design without the necessity of complex conversion
Operating wavelength (nm)
Measuring range (nm)
-70 to +10 @1550nm
-50 to +30 @1550nm
Power meter optical interface
Continuous operating time (h)
>40 (@ liquid crystal backlight off, no red light output)
Power supply mode
Two AA batteries
Automatic off time (min)
Operating temperature (℃)
-10 to +50
Storage temperature (℃)
-20 to +70
165´65´35 (L ´ W ´ H)
Long press to turn on/off the machine. Short press to switch the automatic off function
Liquid crystal backlight key. Short press to switch backlight states
Wavelength selection key. Short press to switch operating wavelengths
Short press to view a reference value. Long press to set a reference value to the current measuring value
Short press to switch display units
Test loss operation:
1. Measure reference signals in the measuring mode.
2. Long press the REF key to save a measuring reference value.
3. A saved measuring value is an attenuation value.
4. View the measuring reference value by short pressing REF.
5. Query about the current actual measuring value by short pressing the unit toggle key ( key).
Press and keys at the same time. The current test wavelength flashes. Enter the self-calibration mode.
Perform up-regulation and down-regulation using key and key respectively. After calibration,
short press key to save a calibration value and finish self-calibration.